How to Use a Multimeter on a Circuit Board

Posted by

Introduction

A multimeter is an essential tool for testing and troubleshooting circuits and devices. It can measure voltage, current, resistance, continuity, and other parameters. Using a multimeter properly requires understanding the different functions, settings, and techniques for making accurate measurements. This guide will cover the basics of using a multimeter to test circuit boards and identify problems.

Multimeter Functions

Multimeters have several functions for making different types of measurements:

Voltage Measurement

The voltage function measures the voltage or potential difference between two points in a circuit. Set the multimeter to the appropriate voltage range for the expected voltage you plan to measure. Touch the black probe to ground and touch the red probe to the test point to measure the voltage.

Current Measurement

The current or amperage function measures the flow of charge through a circuit. You’ll need to connect the multimeter in series with the circuit to take a current measurement. Start with the highest current range and work down until you get to the appropriate range for the actual current.

Resistance Measurement

The resistance function is used to measure the resistance in ohms between two points in a circuit. Make sure power is disconnected from the circuit before making resistance measurements. Touch the probes across the points where you want to measure resistance.

Continuity Test

The continuity test sends a small current through the circuit and checks if a complete path exists between two points. This is used to detect shorts or opens in a circuit. You’ll hear a beeping sound if a complete path exists.

Diode Test

The diode test function applies a voltage to a semiconductor junction to determine if it is functioning properly as a diode by measuring the voltage drop. Touch the black probe to the cathode and red probe to the anode of the diode.

Multimeter Features

• Display: The digital display shows the measurement values and function settings. Some models have an analog bargraph for a quick reading.
• Selection Dial: This dial selects the various functions and ranges for making measurements.
• Inputs: Input jacks allow connection of the test probes to make measurements.
• Hold Button: The hold button freezes the display for convenient reading.
• Backlight: Some multimeters include a backlight to illuminate the display for use in dim areas.
• Auto-Ranging: Auto-ranging automatically selects the appropriate measurement range based on the input. Manual ranging can also be used.

Multimeter Safety

Using a multimeter safely is extremely important, especially when working with high voltage:

• Turn off power to circuits before making resistance or continuity measurements.
• Double check that the correct function and range are selected on the dial for your measurement.
• Do not exceed the maximum input limits.
• Use caution when working with high voltages.
• Make sure the multimeter probes have insulated shafts and finger guards.
• Frequently inspect test leads for damage and insulation breakdown.
• Turn off the multimeter when not in use.

Testing a Circuit Board with a Multimeter

Circuit boards contain various interconnected components. Here are some techniques for using a multimeter to test a circuit board:

Visual Inspection

First, visually inspect the board for any obvious problems like damaged tracks or burnt components. Also look for deteriorated solder joints and correct connection of connectors.

Power Connections

Check the power supply connections to the board using the voltage function. Make sure the voltages are within specification for the board. Also check the ground connections.

Continuity Testing

Use the continuity function to test conductive tracks and connections on the board. Probe along suspect sections of the board to ensure there are no opens.

Component Testing

Test resistors, capacitors, diodes, transistors, and other components for proper function. Compare measured values to expected values based on circuit schematics and board documentation.

Signal Tracing

Trace signals using the voltage function to isolate malfunctioning stages. Work logically from inputs to outputs to pinpoint signal loss.

Isolation Testing

Isolate faulty sections by lifting components and testing interconnections. Repeat until the faulty component or connection is identified.

Step-by-Step Multimeter Usage on a Circuit Board

Follow these steps to systematically use a multimeter to test a circuit board:

1. Deenergize the Board

Turn off and unplug the device so the board has no power. Make sure capacitors are discharged.

2. Visually Inspect the Board

Look for any obviously damaged or defective components or PCB tracks. Also check for poor solder joints and connectors.

3. Check Power Supply Lines

Set the multimeter to the 20V DC voltage range. Place the black probe on ground. Touch the red probe to the board power supply lines like VCC and verify the voltages match expectations.

4. Check for Shorts

Configure the multimeter for the lowest resistance range, 200 ohms or less. Probe across power and ground lines to make sure there are no shorts.

5. Verify Ground Connections

Check ground pins and traces for continuity to the ground plane. There should be no more than a few ohms of resistance.

6. Test Resistance of Components

Consult a schematic and measure resistors, potentiometers, and other components. Compare to expected values.

7. Test Capacitors

Use the continuity function to test that capacitors are not shorted. Discharge fully before testing. Check capacitance if needed.

8. Check Diodes

Utilize the diode test function to confirm diodes are not shorted or open. Make sure they conduct only in one direction.

9. Analyze Circuits and Take Voltage Measurements

Working from inputs to outputs, trace signals and analyze circuits. Take voltage measurements at test points.

10. Isolate Faults

Narrow down faults by lifting components, testing interconnections, and substituting known good components.

11. Verify Fixes

After identifying and replacing faulty components, retest to verify proper board operation.

Tips for Using a Multimeter on a Circuit Board

• Work logically from inputs and power to outputs to isolate faults.
• Confirm proper ground connections which are needed for accurate measurements.
• Compare measurements to schematics and diagrams to identify defects.
• Start on the highest range and work down when measuring an unknown voltage or resistance.
• Check for shorts, opens, and component faults to narrow down problems.
• Look for solder bridges which can cause unwanted connections.
• Keep probes firmly pressed into the test points for accurate low resistance measurements.
• Apply a heat sink clip when testing sensitive semiconductors to avoid overheating.
• Take safety precautions like isolating high voltage points and discharging capacitors.
• Properly store and handle the multimeter to avoid damage which could affect measurements.
• Clean the probe tips occasionally to remove oxidation and material buildup.

Common Circuit Board Faults and Defects

Some typical faults and defects found on circuit boards include:

• Short circuits between tracks or pads
• Open traces or damaged PCB causing open circuits
• Cold solder joints causing intermittent connections
• Overheated or burnt components
• Cracked PCB laminate layers
• Bad solder causing high resistance connections
• Poor wire crimps or connectors
• Failed or out of tolerance resistors
• Shorted or leaky capacitors
• Shorted or open diodes
• Defective transistors
• Faulty integrated circuits
• Power supply line problems
• Ground plane continuity issues
• Erratic voltage regulator operation

Using a multimeter to methodically step through circuits makes it possible to detect and isolate these and other faults.

Troubleshooting No Power Problems

If a device with a circuit board is completely dead, here are some steps to troubleshoot power problems with a multimeter:

• Check AC input voltage to board or power supply
• Verify fuse is good
• Test power switch with continuity check
• Check transformer operation with AC voltage measurements
• Test rectifier diodes on DC side with diode function
• Inspect filter capacitors for bulging or leaking
• Verify voltages on DC power lines
• Check ground connections from board to chassis
• Look for shorts on the board causing protectives circuits to open
• Determine if voltage regulators are working properly

Carefully checking each part of the power system will help identify what component or connection is preventing proper energizing of the board.

Multimeter Limitations and Challenges

While multimeters are versatile tools, they do have some limitations:

• Limited accuracy, especially on inexpensive models
• Not well suited for high frequency AC measurements above a few MHz
• Only display single measurement result at a time
• Manual ranging can be tedious for large sets of tests
• Difficult to connect and take readings in dense, complicated circuits
• Small display and limited displays make interpreting values tricky
• Don’t provide detailed component analysis beyond basic tests

Understanding these limitations allows use of alternative techniques like oscilloscopes, signal analyzers, and dedicated circuit testers when needed.

Choosing a Multimeter for Circuit Board Testing

Here are some factors to consider when selecting a multimeter for troubleshooting circuit boards:

• Accuracy – A multimeter with accuracy of 0.5% or better ensures reliable measurements.
• Resolution – High resolution allows seeing small measurement changes. 5-1/2 digit models offer better precision.
• Voltage rating – Select a model rated for the highest voltages in the circuits to be tested.
• Current rating – Choose a multimeter that handles the maximum expected circuit currents without blowing an internal fuse.
• Resistance range – A low ohm range down to 0.1 or 1 ohm is useful for checking continuity.
• Capacitance range – The ability to measure capacitance can help testing capacitors.
• Frequency range – Some boards operate at higher frequencies requiring 20 MHz bandwidth or more.
• Safety rating – For mains voltages, a CAT III or better safety rating is advised.
• Features – Considerations like auto-ranging, PC interface capability, graphical display, and recording functions.

Selecting a multimeter with appropriate specifications and features will maximize efficiency and capabilities when testing circuit boards.

For complex issues, some additional techniques can help troubleshoot circuit boards:

• Signal injection – Input test signals at key points to trace operation.
• Oscilloscope analysis – View signals and timing to detect glitches.
• Circuit simulation – Model the circuit for detailed analysis.
• PC tools – Use PC-based diagnostic tools specific to the board or components.
• Thermal imaging – Detect hot spots pointing to components drawing excessive current.
• Signal tracing – Electronic probes allow tracing signals deep inside complex ICs.

These and other methods complement multimeter measurements to pinpoint faults. Combining different troubleshooting tools yields the fastest results.

Conclusion

Knowing how to properly utilize a multimeter is critical for testing and debugging circuit boards. Follow solid procedures, make systematic measurements, and compare to expected values. With experience, analysis skills will improve leading to faster isolation of malfunctions. A quality multimeter along with good probing techniques, patience, and persistence will uncover most board faults.

Tag Cloud

There’s no content to show here yet.